Labview Error Codes Visa
User-friendly, computer-controlled instrumentation and data analysis techniques are revolutionizing the way measurements are being made, allowing nearly instantaneous comparison between theoretical predictions, simulations, and actual experimental...https://books.google.com/books/about/LabVIEW_based_Advanced_Instrumentation_S.html?id=TTP1KDSzMg4C&utm_source=gb-gplus-shareLabVIEW based Advanced Instrumentation SystemsMy libraryHelpAdvanced Overview Instrument Driver Technology: Plug and Play Supported Language(s): LabVIEW 6.1 Supported Regional Settings: English and Universal desktops (period or comma separator for number decimal point) Supported Models: Lambda Genesys Series DC Code Description −1073807360 Unknown system error (miscellaneous error). −1073807346 The given session or object reference is invalid. −1073807345 Specified type of lock cannot be obtained, or specified operation cannot be performed, Checksum Message Checking All messages sent to the Genesys power supply have a checksum added to the end of the message. http://edsdefence.com/labview-error/labview-error-codes-2011.php
Danny Diaz01-21-2006, 04:06 PMSeriously, this is getting way too complicated and our team is about to give up on the camera again. Verify that the command being sent to the serial device has the termination character that the device requires. Re-run the main application. You're absolutely correct, VISA is a Virtual Instrument Software Architecture (VISA) capable of programming, configuring, and troubleshooting a wide range of instrumentation systems (that, coincidentally enough, can use a serial interface
Labview Visa Error -1073807339
Before changing anything I did confirm that the device was displaying the REM indicator on the physical front panel during my previous failures. Attempt to program a setting outside of the model rating EXAMPLES: DRIVER ERRORS Decimal Code Hex Code Meaning -1074003951 BFFC0011 No response from "ID Query" in "Initialize.vi". And I know that NI MAX can communicate using VISA serially.
We set the port to the correct baud rate, wired the camera, turned everything on, etc. Configuring the Instruments and Drivers The controls on the "Initialize.vi" driver must match the settings on the power supply you want to communicate to. You can then modify this example to meet your specific needs. Labview Error 200279 Unsupported Features 11.
However, you are able to set this property/attribute at any point during the session, allowing you to configure it as needed. Daqmx Error Codes Do not invoke any other handlers on this session for this event. 1073676441 Operation completed successfully, and this session has nested shared locks. 1073676442 Operation completed successfully, and this session has A character was not read from the hardware before the next character arrived. −1073807250 The path from trigSrc to trigDest is not currently mapped. −1073807248 The specified offset is not properly Sumathi, P.
Why does this happen? Labview Custom Error Codes Please tell us why. Now accompanied by LabVIEW 2011, this classic second edition, focusing on LabVIEW 8.0, delves deeply into the classic features that continue to make LabVIEW one of the most popular and widely Please Contact NI for all product and support inquiries.
Daqmx Error Codes
Related Links: KnowledgeBase 0Q0H595Q: Can VISA Communicate with Parallel Ports?KnowledgeBase 18QGBH0O: Setting Individual Bits for the Parallel Data Port Using LabVIEWKnowledgeBase 3B8H2CL6: How Can I Verify That My Parallel Port is Your cache administrator is webmaster. Labview Visa Error -1073807339 Examples: see "RESPONSE ERRORS" table below The VISA standard allows custom error codes in the range: -1073999873 to -1074001919 ( BFFC0801 to BFFC0FFF hex) The instrument errors are formulated in "GENser Labview Error 1073807339 Poor|Excellent Yes No Document Quality?
Accessing the Driver in LabVIEW 5. check my blog Addressing the Instrument 8. Must have "?" in query string -1074001896 BFFC0818 After command, received a response but it was not "OK" or any valid error code -1074001888 BFFC0820 Checksum error: did not verify CS Addressing the Instrument Every driver has a "GEN Addr" control terminal. Labview Error Daqmx
Please see below for instruction on setting the VISA Timeout:LabVIEW: You can use VISA Set Timeout located in the Functions palette under Instrument I/O » VISA » VISA Advanced LabWindows/CVI: You The java CMUCAM2 program from 2005 works--we can grab frames, control servos, etc. Preview this book » What people are saying-Write a reviewLabviewbased Advanced Instrumentation SystemsUser Review - Overstock.comThis is a technical book and not intended for Entertainment unless you enjoy this sort of this content My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
For more details, reference the LabVIEW Help under: Help » VI, Function, and How-To Help. 6. Error Labview Fpga Known Issues 13. YourFeedback!
This could be caused by a number of things, like: Connecting the CMUCam2 to both the RC and the PC at the same time.
Unsupported Features The following Genesys capabilities are not supported by these drivers: - Event and Enable registers (FEVE, FENA, SEVE and SENA commands) - SRQ interrupts - Calibration commands - Calibration Installation Instructions Obtain a copy of the driver software. Click the "Browse" button under "Open" 3. Labview Fpga Error Code You may not have the proper driver software installed.
Attempt to turn the output ON while a fault is keeping the output OFF -1074001407 BFFC0A01 Command error "C01". Refer to the KnowledgeBase for more information about correcting errors in LabVIEW. This time both the viWrite and viRead execution worked, resulting in the buffer message "Keithley Instruments Inc., Model 2611A, 1339183, 2.1.6\n".3) NI LabVIEW: (fail) I then tried a simple VISA read/write http://edsdefence.com/labview-error/labview-error-codes-list.php You may have a bad serial cable.Try another serial cable. 5.
See the Genesys Technical Manual, Chapter 7, for details on setting the address and Baud rate for each instrument and for selecting and connecting the RS-232 or RS-485 cables.