Labview Error Code 30
There is a size mismatch between the data in the reference and the data wired to the Data Value Reference Write Element border node. 1586 Compiled code is out of date. For example, a device-level ibwrt returns EBUS if it is unable to send the addressing command bytes. 15 Serial poll byte queue overflow. 16 SRQ stuck in ON position. The palette view format must be Icons, Icons and Text, or Text. If it is reentrant, use the Open VI Reference function with the options input set to 0x08 to prepare the VI for reentrant run or use the Open VI Reference function weblink
Verify that the VI is marked to save its compiled code separately and that it is not broken. Code Description −2147467259 Unspecified error. −1967345152 Invalid refnum. To correct this error, you must obtain a valid license for the VI and its containing library. 1389 You are attempting to save or copy a VI that is either in For example, hexadecimal notation is not a valid format for floating point numbers. 1434 The precision is greater than the maximum allowed value for this format. 1436 Numeric precision cannot be https://forums.ni.com/t5/LabVIEW/Why-do-i-get-Error-Code-30-after-recompile-of-working-code/td-p/3178798
Verify the values you wired to the Open VI Reference function. 1097 An exception occurred within the external code called by a Call Library Function Node. If you must use device names for your instruments, then make sure they are correctly configured in the Device Templates of the GPIB Configuration Utility (see your NI-488.2 User Manual for To correct this error, load an instance of the polymorphic VI rather than the polymorphic VI itself. 1554 The current LabVIEW target cannot load control VIs. 1556 The reference is invalid. The front panel must already be open before you close it or set its state. 1150 Cannot open a front panel that is already open.
Cart|Help You are here:NI Home > Support > Manuals > LabVIEW 2012 Help General LabVIEW Error Codes »Table of Contents LabVIEW 2012 Help Edition Date: June 2012 Part Number: Try switching cables to verify that the cable is not broken. Save any work to a new location and restart LabVIEW. This error occurs when a driver function is called with an invalid parameter. 5 Command requires GPIB Controller to be System Controller.
Note: If the following steps do not remedy the issue, please contact NI support. For example, the network connection is down or a network cable is unplugged. 7 File not found. Refer to the KnowledgeBase at ni.com if you receive this error in one of the following situations: while using the VI Server to call a VI dynamically or while building an http://forums.ni.com/t5/LabVIEW/I-get-a-message-like-quot-labview-load-error-code-30-could-not/td-p/32950 Attempted to read from or write to a file of a version incompatible with the write/read function version.
Refer to the National Instruments Web site to download the VIs. 1196 Cannot list the same terminal more than once in the grown region of the expandable subVI. 1197 This operation You must create these objects in the LabVIEW development system. 1124 VI is not loadable. The file might have been moved or deleted, or the file path might be incorrectly formatted for the operating system. There is already a VI in this hierarchy with this name. 1449 Arrays must have at least one dimension and a maximum of 63 dimensions. 1450 One or more untitled library
Select File»Open to open the VI and verify that you can run it. hop over to this website This error occurs when you try to use the Active Frame property with a Conditional Disable structure. mikeporter Proven Zealot 08-14-2015 06:53 PM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Try looking below the "or" There are not enough format specifiers to match all of the arguments of a Format Into String or Scan From String function. 84 Too many format specifiers.
You cannot get a new reference to the application instance or load a VI in the application instance. 1533 A shared variable node cannot be created from the specified terminal. 1534 have a peek at these guys Solutions: Give the GPIB board System Controller capability by calling ibrsc 1 or by using the GPIB Configuration Utility. Use the Window Appearance page to configure if the front panel is shown when the VI is run or loaded. The input string does not contain data in the expected format.
Copy this new Alphanumeric ID. Another possible cause for this error is there might be a bad network connection. The sound format of this file is not recognizable. check over here Use the Get Volume Info function to return the proper sector size.
ENOL (2)Error Condition: Function detected no Listener(s).Description: GPIB communications require a single Talker (to write data messages) and one or more Listeners (to read data messages). The requested entry could not be found. −120000 LabVIEW SignalExpress is required to open this ExpressVI. For example, the network connection is down or a network cable is unplugged. 8 File permission error.
Save the new VI or function to a different directory than vi.lib so you do not overwrite the original. 44 Object ID too low. 45 Object ID too high. 46 Object
tcarrico Member 08-14-2015 04:32 PM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Hi, I'm seeing Error code Check printer configuration. 1016 Cannot load History. 1017 VI has been modified on disk since it was last loaded or saved. 1018 Unspecified error occurred. 1019 One or more untitled subVIs To correct this error, save to a folder instead of an LLB. 1497 Cannot unlock a library for edit when instances in running VIs exist. 1498 Library has errors. This error occurs when you try to use the Default Frame property with a Diagram Disable structure.
This error is only returned by functions that indirectly send out command bytes. This problem happens when the board is not physically plugged into the system, the I/O address specified during configuration does not mach the actual board setting, there is a system conflict A bad format specifier was found in the Format String input to a Format Into String or Scan From String function. 83 Too few format specifiers. this content The sound driver or card cannot accommodate the specified configuration.
Use the Get Volume Info function to return the proper sector size. The front panel must already be open before you close it or set its state. 1150 Cannot open a front panel that is already open. You may expect to see EOI (end or identify, one of the five bus management lines) as the termination method, but if the instrument does not set the EOI line when or @.
You must create these objects in the LabVIEW development system. 1124 VI is not loadable. Another application is currently using the device. EARG (4)Error Condition: Invalid argument to a function call.Description: EARG results when an invalid argument is passed to a function call. This error occurs when you try to use the Active Frame property with a Conditional Disable structure.
There are duplicated items in the array. 1304 The array index is outside of the array bounds. 1305 The required page cannot be found. 1306 Unable to load new code resource In this case you must rebuild the application for the target OS and make sure SSE compiler settings in the build specifications match the target platform. If you have re-connected a removable device, you may need to restart LabVIEW to use the device again. 29 Print aborted. 30 Bad print record. 31 Print driver error. 32 Operating Please try the request again.
To correct this error and inline the subVI, remove the implicit control reference. 1485 LabVIEW cannot inline the subVI into its calling VIs because the block diagram of the subVI contains Double-click the VI or function and replace the general error handling with an error out indicator or some other form of error handling. Please tell us why. The specified privilege ID is out of range. −1967345149 Invalid access type.
Navigate to the VI for the reference and click the Open button to configure the reference. 1190 The operation is not allowed when the control has key focus. 1191 The wire This troubleshooting information is continued in Knowledge Base 1XOHEPPH: GPIB Error Codes and Common Solutions (Part 2)and Knowledge Base 3CO9NH3F GPIB Error Codes and Common Solutions (Part 3).